3L/min Adjustable Oxygen Machine for Home 301W
Subcribe to back in stock notification
customers are viewing this product
USA Address: 3-5 Business Days
Other Address: 7-10 Business Days
- 1.Molecular sieve: high-efficiency lithium sieve, strong adsorption, high oxygen rate, strong stability.
- 2. Oxygen concentration: medical standard, when the flow rate in 1-3L, oxygen concentration stable at 93% ± 3%, continuous and stable supply of oxygen
- 3. Compressor: compressor with overpressure safety valve and thermal protector, to better ensure the safety of the compressor and the machine, pure copper oil-free air double-cylinder compressor, small energy consumption, strong power, low noise, fast heat dissipation
- 4. Eight-stage filtration upgrade system: no impurities, pure oxygen. Every filter is a filter of various impurities, particles, molecular sieve nitrogen, and oxygen separation, to ensure that the oxygen is out of the clean.
- 5. Intelligent Control: IMD's beautiful large color panel, easy to operate, large color LED screen, Flow regulation, voice broadcast function, timer operation function, and infrared remote control, let it be easier and more convenient.
- 6. Low noise: the measured noise is as low as 43.6 dB, and the oxygen concentration is high, which can still be maintained in continuous operation
- 7. 2in1 Oxygen in-ha-la-tion to-mi-zation: Special interface for ato-miza-tion, Oxygen generation, and ato-miza-tion are carried out at the same time
- 8. Small size: lightweight of about 10.8kg, light, and portable, oxygen intake anywhere, anytime!
- Power supply: AC110V, AC220V
- Output power: 300VA
- Oxygen concentration: 90%(Adjustable by flow)
- Oxygen flow: 0.5-3L/min
- Net weight: 10.8kg
- Outline size: 320x210x415mm
- Safety system: Current overload protection, high temperature of compressor, shutdown
- Electrical classification: Ⅱ,Type BF
- Operating mode: continuous
- Ambient Temperature: 5℃～40℃
- Relative humidity: ≤80%
- Atmosphere pressure: 86kPa~106kPa
Note: When storage temperature is lower than 5C, the equipment be placed at normal operating temperature for 4 hours before